ATTEN ADSCAL vs Rigol DSe were the two options left for considering . Manual, tracking, automatic cursor measurement function;. I have recently bought an ATTEN ADSCAL+ oscilloscope. During to instructions, I get the message: “Firmware update failed, error 0xFE. Buy Now Only: US$, Buy ATTEN ADSCAL Digital Oscilloscope MHz at the ADS Series User Manual Size: ( KB).
|Published (Last):||14 November 2013|
|PDF File Size:||14.55 Mb|
|ePub File Size:||10.12 Mb|
|Price:||Free* [*Free Regsitration Required]|
Application Of X-y Function 3. Brand new with one year USA warranty. Turn off the Digital Filter function. There are many bugs and glitches, and the manufacturer will not update the firmware. Save setup to USB flash drive, follow these steps For example: Sync Line Num All lines Select appropriate video sync. Don’t show me this message again. Print Setup Enter the print setup menu to set print options. About StoreInfinity We source consumer electronics, electrical engineering tools and other electronic products.
General safety summary Avoid Exposed Circuitry. The shapes with two arrows beside them represent the front panel knobs. For example, you might look at waveform amplitude and determine that it is a little more atyen mV. X-y Format adjust grid brightness.
Atten ADSCAL Oscilloscope Product Reviews
Page 15 Picture 4. This Chapter shows you how to operate following tasks: You can view the same waveform in XY mode. Positive pulse width larger than pulse width setting Positive pulse width equal to pulse width setting Negative pulse width less than pulse adz1102cal setting Page 54 Picture Operate Instruction: I use a ‘scope all the time.
You cannot adjust the horizontal position and scale of the reference waveform.
ATTEN ADS1102CAL Digital Oscilloscope 100MHz
If a product or CRT proves defective within the respective period, ATTEN will provide repair or replacement as described in the complete warranty statement. The oscilloscope takes a few seconds to capture the screen image.
Picture Note X-Y mode waveforms are not stored as reference waveforms. Option Setting Instruction Invert Turn on invert function. ADS series digital storage oscilloscopes combine mature and steady hardware configuration, high-powered digital signal processing technology and functionality concepts with a mainstream display interface and the operation panel.
If there is no signal wave in the screen after gathering the signal, please do as following steps: In these cases, you should use the Peak Detect mode to acquire data. Load Recall the Factory setup. Table Set trigger mode to slope trigger function menu 1 Option Setting Instruction Type Alternative The trigger signal comes from two vertical channels das1102cal you use alternative trigger. One picture is worth a thousand words; this ads110c2al a now a detailed, picture album.
Set trigger type of the vertical channel signal to Mode Pulse Pulse trigger. Select operating object Create new needed folder Delete present folder Picture Select operating object Create new files in present directorys Delete present file Recall stored files or present settings Picture Modify the file name Return to the last menu Picture In this module,Setting stores to File,Waveform stores to File,storing pictures and CSV stores to USB flash drive all have this function.
Page 40 According to the tested options and source speciality, make sure the window you need to use. Taking Cursor Measurements 3. Online Help Function 2.
When you recall the setup, the oscilloscope will ayten in the mode from which the setup was saved. Add to Wishlist Add to Compare Share.
The following contents are the brief introduction for the front panel function, which is useful to be familiar with the operation of the ADS series Digital Storage Oscilloscope in short time.
Pop-up Menu display pattern is more convenient for users attem it. Probe attenuation factor Status changes with the probe attenuation Picture 5.
Continuously acquires waveforms or stops the acquisition.
Classical, Modern, Tradition, Succinct. Taking Manuzl Measurements 3. I used them back ‘in the day’ at IBM. Accidence of the front panel and user interface